Mass spectrometry is an analytical technique used to identify the chemical composition of a sample. The method typically works by bombarding a sample with electrons and breaking it into charged molecules or molecule fragments. These ionized particles are then separated, normally by accelerating them in an an electromagnetic field, and the mass to charge ratio for each compound is then determined. To detect the ionized particles an electron multiplier is normally used and the results are displayed as a spectra which plots the number of detected ions as a function of their mass to charge ratio. The electron multiplier plays a crucial role in detecting the ions and producing the electrical signals that need to be analyzed.
For situations that require fast and accurate capture of the electrical signals Spectrum has a range of digitizers that can be employed. For example, 8 bit digitizers are available that offer from 1.25 GS/s up to 5 GS/s sampling rates and 1.5 GHz bandwidth. These products are ideal for capturing and analyzing pulses that go down into the nano-second and sub-nanosecond region. The ultra-fast sampling rates also ensure excellent timing resolution in time-of-flight mass spectrometry systems (TOF-MS). When larger dynamic range and more sensitivity is required high-resolution 14 and 16 bit digitizers are available with sampling rates from 130 MS/s up to 500 MS/s. These high-resolution products deliver outstanding signal-to-noise ratio's (up to 72 dB) and spurious free dynamic range (of up to 90 dB) so that even the smallest signals can be detected and analyzed.
Small and compact the digitizers can be installed directly in any free PCIe slot in most modern PC's. They are equipped with ultra-fast trigger circuits so that the dead-time between acquisitions can be extremely small (down to as little as 16 ns). Together with large on-board memories (up to 4 GSamples/card) and advanced streaming and readout modes this makes the digitizers ideal for applications where acquisitions need to be made over long periods of time and where high trigger rates may be encountered. Data can be stored in the on-board memory or streamed in FIFO mode over the fast PCIe bus of the digitizer to a PC at rates up to 3.4 GB/s. To help with data analysis and data reduction Spectrum's M4i series of digitizers also feature on-board FPGA based processing functions that can be perform on-the-fly Averaging and Peak detection routines. Averaging a signal helps remove unwanted noise and improves sensitivity while Peak Detection can be used to help locate pulses and provide time interval analysis.
Each digitizer card can have from one to four channels and up to eight cards can be linked together with Spectrum's StarHub system to create instruments with up to 32 fully synchronous channels. The flexibility allows you to match your system perfectly whether it be for use with a single detector or for applications where multiple detectors and detector arrays are deployed.
To allow easy integration into mass spectrometry systems Spectrum makes available software and instrumentation drivers that work with 32 bit and 64 bit versions of Windows and Linux. Programming of the cards is possible using a wide range of languages such as LabVIEW, LabWindows/CVI, C++, MATLAB, Borland Delphi, Visual Basic, VB.NET, C#, J# and IVI.
Typical mass spectrometry applications include chemical analysis and life science measurements using TOF-MS (including MALDI-TOF), accelerator mass spectrometry (AMS), inductively coupled plasma mass spectrometry (ICP-MS) and thermal ionization mass spectrometry (TIMS). Mass spectrometry can also be combined with chromatographic compound separation methods such as liquid chromatography (LC-MS), gas chromatography (GC-MS) and ion mobility mass spectrometry (IM-MS).
Spectrum Product Features
- High Sampling Rates up to 5 GS/s and 1.5 GHz bandwidth
- 14 and 16 bit Resolution
- Fast Trigger and Read-Out Rates
- External Clock and Reference Inputs
- FPGA based Block Average and Block Statistics (Peak Detect) Options
Matching Card Families
Using software based fast block averaging
The block, or segmented memory, averaging mode is used with Digitizers for different applications where incoherent noise needs to be removed from a signal. Independent of the manufacturer of the digitizer all FPGA based hardware implementations of the block averaging mode limit the maximum size of the segment to be averaged. The limit depends on the capacity of the FPGA and usually ranges from 32k up to 500k samples.Read more >>
Case Study: REMPI Mass Spectrometry
The scientific field of Mass Spectrometry (MS) has been under constant research and development for over a hundred years, ever since scientists discovered that charged particles could be separated via electromagnetic fields based on their charge-to-mass ratio (Q/m). While spectrometry systems now come in numerous configurations their performance largely depends on a few fundamental elements. In the modern Mass Spectrometer this is normally the ion source (for exciting ions from the material under analysis), the mass analyzer, the particle detector and its associated electronics.Read more >>
Mass Spectrometry and the Modern Digitizer
The scientific field of Mass Spectrometry (MS) has been under constant research and development for over a hundred years, ever since scientists discovered that charged particles could be separated via electromagnetic fields based on their charge-to-mass ratio (Q/m). While spectrometry systems now come in numerous configurations their performance largely depends on a few fundamental elements.Read more >>
Signal Processing for Digitizers
Modular digitizers allow accurate, high resolution data acquisition that can be quickly transferred to a host computer. Signal processing functions, applied in the digitizer or in the host computer, permit the enhancement of the acquired data or the extraction of extremely useful information from a simple measurement.Read more >>
- At the Division of Advanced Manufacturing, Tsinghua University, China, they are developing ways to lower the cost of Dual-comb Spectroscopy systems. A new approach uses an M4i.4420-x8 250 MS/s, 16-bit Digitizer to acquire signals and passes the data to a computer where it is analysed using a simple frequency domain post-processing method. A white paper explaining their system and the process can be found here
- The Department of Physics, Keio University, Japan is using an M4i.2230-x8 5 GS/s Digitizer for high speed data acquisition with fast time domain averaging to perform scan-free high-resolution direct-comb spectroscopy where the mode spacing of an optical frequency comb is reduced down to 260 kHz by phase modulation. Details on the development can be found here
- The Oklahoma State University, USA, has developed a system and method that improves the quality of step-scan Fourier Transform Infrared spectroscopy data by improved removal of dark voltage. The experimentation uses a digitizerNETBOX DN2.441-02 that can acquire signals by sampling them at rates up to 130 MS/s with 16 bit resolution. A patent application outlining the scope of the work is available here
- The Department of Chemistry, National Taiwan University, in Taiwan are studying the roaming dynamics and conformational memory in photolysis of formic acid. They use an M4i.4410-x8, 130 MS/s, 16 bit, Digitizer for data acquisition in their Time-resolved Fourier-transform Infrared Emission Spectroscopy system. A research paper on the topic can be found here
- At the Technical University of Dortmund, Germany, they are using a 5 GS/s M4i.2234-x8 Digitizer in nonstationary quantum state tomography, adapting the technique to the special requirements of ultrafast spectroscopy. A white paper on the topic can be found here
- At the Norwegian University of Science and Technology they are using an eight channel M2p.5946-x4 80 MS/s, 16 bit, digitizer card as part of their studies into biomedical laser spectroscopy. The development of a Mid-Infrared Laser Spectroscopy for Glucose Sensing is discussed here
- The Institute of Optics and Electronics, Chinese Academy of Sciences, Chengdu, China, uses both a Spectrum 14 bit AWG M2i.6021 and 12 bit digitizer M2i.3010 as key elements in their experimental Optical Feedback Cavity Ring-Down (OF-CRD) system. Learn how signal processing can help extract low signal-to-noise ratio signals by clicking here
See how a Spectrum M2i.4963-exp digitizer card, programmed under LabVIEW, is used at the Department of Applied Physics and Electronics, Umeå University, Umeå Sweden, in an EC‑QCL (external cavity – quantum cascade laser) system for real‑time breath gas analysis of CO and CO2 by clicking the link here
At the University of Konstanz a fast high resolution Spectrum M3i.4142 digitizer provides nanosecond time-resolution in an FTIR spectroscopy system that’s used to investigate protein-membrane interactions. See the full details here
Electron Spin Resonance (ESR) is a key technique for the study of the structure and dynamics of molecular systems. At the Walter Schottky Institute of the Technical University Munich they are using a Spectrum M4i.4451-x8 digitizer card to help detect, acquire and analyze spin echoes in a pulsed ESR system. A white paper summarizing the experimental setup and results is available here